- •Significantly smaller peak force during nail extraction with a radius of curvature of 1.0 m
- •No difference in peak force between 1.0 m and 1.5 m radius of curvature while implant extraction
- •Plastic deformation was observed in all femur nails post extraction
The impact of the nail radius of curvature, as one of the most important design features in modern femoral nails on the ease of nail removal, remains unknown. Therefore, the aim of this study was to investigate force, energy, and nail deformation of different nail designs.
Nail insertion and extraction was performed on six pairs of fresh-frozen human cadaveric femora on a material testing machine with two different nail systems – Trochanter femoral nail ADVANCED™ Nailing System with a radius of curvature of 1.0 m and Proximal Femoral Nail Antirotation System with a radius of curvature of 1.5 m. Deformation was measured after insertion (plastic and elastic deformations) and extraction (plastic deformations).
The peak force during nail removal was significantly lower in the first group (274.5 ± 130.4 N) compared to the second group (695.2 ± 158.8 N, p = 0.001). Plastic deformation was observed in all implants, being significantly larger in the Proximal Femoral Nail Antirotation System (p = 0.027). There was a strong positive correlation between the first peak force during nail removal and nail insertion (r = 0.802, p = 0.002) as well as between extracting energy and insertion energy (r = 0.943, p < 0.001).
The results from this study showed that a radius of curvature of 1.0 m is easier to remove from the set of cadaver femora. Furthermore, our findings support the idea of further reducing the nail radius of curvature below 1.0 m in order to more closely match the anatomy of populations with strong-bowed femora.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Clinical Biomechanics
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Cortical encroachment after cephalomedullary nailing of the proximal femur: evaluation of a more anatomic radius of curvature.J. Orthop. Trauma. 2013; 27: 303-307
- Is routine removal of intramedullary nails justified.Injury. 1999; 30: 79-81
- Difficulty in removal of a femoral intramedullary nail: the geometry of the distal end of the nail.Yonsei Med. J. 2003; 44: 1083-1086
- Incidence of intramedullary nail removal after femoral shaft fracture healing.Can. J. Surg. 2007; 50: 13-18
- Removal of intramedullary nails from the femur: a review of 45 cases.J. Orthop. Trauma. 1996; 10: 560-562
- Removal of intramedullary rods after femoral shaft fracture.J. Orthop. Trauma. 1992; 6: 460-463
- Quantification of cephalomedullary nail fit in the femur using 3D computer modelling: a comparison between 1.0 and 1.5m bow designs.J. Orthop. Surg. Res. 2016; 11: 53
- Difficulty in removal of certain intramedullary nails.Clin. Orthop. Relat. Res. 1997; : 202-206
- Intramedullary femoral nailing: removing the nail improves subjective outcome.Injury. 2002; 33: 247-249
- Safe, rapid, and effortless femoral nail removal using a new third-generation universal femoral nail extraction tool.J. Orthop. Sci. 2006; 11: 626-627
- A cadaveric biomechanical study comparing the ease of femoral nail insertion: 1.0- vs 1.5-m bow designs.Arch. Orthop. Trauma Surg. 2017; 137: 663-671
Published online: February 10, 2023
Accepted: February 8, 2023
Received: October 12, 2022
© 2023 Published by Elsevier Ltd.