- •Condyle-spanning construct may lower risks of refracture in osteoporotic femurs.
- •Multiple condylar screws avoided strain elevation caused by distal load transfer.
- •Combined testing and modeling protocol for pre-clinical femoral implant evaluation
Condyle-spanning plate-screw constructs have shown potential to lower the risks of femoral refractures after the healing of a primary Vancouver type B1 periprosthetic femoral fracture. Limited information exists to show how osteoporosis (a risk factor for periprosthetic femoral fractures) may affect the plate fixation during activities of daily living.
Using total hip arthroplasty and plate-implanted finite element models of three osteoporotic femurs, this study simulated physiological loads of three activities of daily living, as well as osteoporosis associated muscle weakening, and compared the calculated stress/strain, load transfer and local stiffness with experimentally validated models of three healthy femurs. Two plating systems and two construct lengths (a diaphyseal construct and a condyle-spanning construct) were modeled.
Osteoporotic femurs showed higher bone strain (21.9%) and higher peak plate stress (144.3%) as compared with healthy femurs. Compared with shorter diaphyseal constructs, condyle-spanning constructs of two plating systems reduced bone strains in both healthy and osteoporotic femurs (both applying ‘the normal’ and ‘the weakened muscle forces’) around the most distal diaphyseal screw and in the distal metaphysis, both locations where secondary fractures are typically reported. The lowered resultant compressive force and the increased local compressive stiffness in the distal diaphysis and metaphysis may be associated with strain reductions via condyle-spanning constructs.
Strain reductions in condyle-spanning constructs agreed with the clinically reported lowered risks of femoral refractures in the distal diaphysis and metaphysis. Multiple condylar screws may mitigate the concentrated strains in the lateral condyle, especially in osteoporotic femurs.
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Clinical Biomechanics
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Epidemiology of periprosthetic fracture of the femur in 32 644 primary total hip arthroplasties: a 40-year experience.Bone Jt. J. 2016; 98B: 461-467https://doi.org/10.1302/0301-620X.98B4.37201
- Standard specification for wrought 18Chromium-14Nickel-2.5Molybdenum stainless steel Bar and wire for surgical implants (UNS S31673).ASTM Int. 2013; 1–6https://doi.org/10.1520/F0138-19.2
- Contribution of muscular weakness to osteoporosis: computational and animal models.Clin. Biomech. 2005; 20: 984-997https://doi.org/10.1016/j.clinbiomech.2005.05.018
- Correlation of postural balance and knee muscle strength in the sit-to-stand test among women with and without postmenopausal osteoporosis.Osteoporos. Int. 2013; 24: 2007-2013https://doi.org/10.1007/s00198-013-2285-x
- Isolated locked compression plating for Vancouver type B1 periprosthetic femoral fractures.Injury. 2009; 40: 1180-1186https://doi.org/10.1016/j.injury.2009.02.017
- Development of axial compression and combined axial compression and torque loading configurations to reproduce strain in the implanted femur during activities of daily living.J. Biomech. 2021; 120110363https://doi.org/10.1016/j.jbiomech.2021.110363
- Impact of periprosthetic femoral fracture fixation plating constructs on local stiffness, load transfer, and bone strains.J. Mech. Behav. Biomed. Mater. 2022; 125104960https://doi.org/10.1016/j.jmbbm.2021.104960
- Simplified mechanical tests can simulate physiological mechanics of a fixation construct for Periprosthetic femoral fractures.J. Biomech. Eng. 2022; 144: 1-9https://doi.org/10.1115/1.4052372
- Evaluation of the reliability of lower extremity alignment measurements using EOS imaging system while standing in an even weight-bearing posture.Sci. Rep. 2021; 11: 1-10https://doi.org/10.1038/s41598-021-01646-z
- UKPMC funders group BoneJ : free and extensible bone image analysis in ImageJ.Bone. 2010; 47: 1076-1079https://doi.org/10.1016/j.bone.2010.08.023.BoneJ
- The unified classification system (UCS): improving our understanding of periprosthetic fractures.Bone Jt. J. 2014; 96 B: 713-716https://doi.org/10.1302/0301-620X.96B6.34040
- Risk factors for periprosthetic femoral fracture.Injury. 2007; 38: 655-660https://doi.org/10.1016/j.injury.2007.02.049
- Periprosthetic femoral re-fractures pathogenesis, classification, and surgical implications.Injury. 2020; 1–7https://doi.org/10.1016/j.injury.2020.11.030
- Osteoporosis and vitamin-D deficiency among postmenopausal women with osteoarthritis undergoing Total hip arthroplasty.J. Bone Jt. Surg. - Ser. A. 2003; 85: 2371-2377https://doi.org/10.2106/00004623-200312000-00015
- Experimentally reduced hip abductor function during walking: implications for knee joint loads.J. Biomech. 2009; 42: 1236-1240https://doi.org/10.1016/j.jbiomech.2009.03.021
- Total hip arthroplasty in patients with osteoporosis.HIP Int. 2020; 30: 370-379https://doi.org/10.1177/1120700019883244
- Comparison of QCT-derived and DXA-derived areal bone mineral density and T scores.Osteoporos. Int. 2009; 20: 1539-1545https://doi.org/10.1007/s00198-008-0820-y
- Prevalence of osteoporosis in osteoarthritic patients undergoing Total hip or Total knee arthroplasty.Arch. Phys. Med. Rehabil. 2008; 89: 2373-2374https://doi.org/10.1016/j.apmr.2008.06.007
- Total hip prostheses in standing, sitting and squatting positions: an overview of our 8 years practice using the EOS imaging technology.Open Orthop. J. 2015; 9: 26-44https://doi.org/10.2174/1874325001509010026
- The biomechanical effect of bone quality and fracture topography on locking plate fixation in periprosthetic femoral fractures.Injury. 2015; 46: 213-217https://doi.org/10.1016/j.injury.2014.10.060
- Risk factors for failure after treatment of a periprosthetic fracture of the femur.J. Bone Jt. Surg. - Ser. B. 2006; 88: 26-30https://doi.org/10.1302/0301-620X.88B1.17029
- The excess mortality due to periprosthetic femur fracture. A study from the Swedish national hip arthroplasty register.Bone. 2007; 40: 1294-1298https://doi.org/10.1016/j.bone.2007.01.003
- The incidence of osteopenia and osteoporosis in women with hip osteoarthritis scheduled for cementless total joint replacement.Bone. 2007; 40: 1041-1047https://doi.org/10.1016/j.bone.2006.11.013
- A review of Periprosthetic femoral fractures associated with Total hip arthroplasty.Geriatr. Orthop. Surg. Rehabil. 2012; 3: 107-120https://doi.org/10.1177/2151458512462870
- Retrospective population-based cohort study of incidence, complications, and survival of 202 operatively treated Periprosthetic femoral fractures.J. Arthroplast. 2021; 36: 2591-2596https://doi.org/10.1016/j.arth.2021.02.060
- Treatment of periprosthetic femur fractures around a well-fixed hip arthroplasty implant: span the whole bone.Arch. Orthop. Trauma Surg. 2014; 134: 9-14https://doi.org/10.1007/s00402-013-1883-6
- Trabecular bone modulus-density relationships depend on anatomic site.J. Biomech. 2003; 36: 897-904https://doi.org/10.1016/S0021-9290(03)00071-X
- Implications of increased medio-lateral trunk sway for ambulatory mechanics.J. Biomech. 2008; 41: 165-170https://doi.org/10.1016/j.jbiomech.2007.07.001
- The impact of hip implant alignment on muscle and joint loading during dynamic activities.Clin. Biomech. 2018; 53: 93-100https://doi.org/10.1016/j.clinbiomech.2018.02.010
- Biomechanical comparison of 2 different locking plate fixation methods in Vancouver B1 Periprosthetic femur fractures.Geriatr. Orthop. Surg. Rehabil. 2011; 2: 51-55https://doi.org/10.1177/2151458510397609
- Re-fractures after periprosthetic femoral fracture: a difficult to treat growing evidence.Injury. 2018; 49: S43-S47https://doi.org/10.1016/j.injury.2018.09.045
- A broader strategy for osteoporosis interventions.Nat. Rev. Endocrinol. 2020; 16: 333-339https://doi.org/10.1038/s41574-020-0339-7
- Muscle strength and body composition are clinical indicators of osteoporosis.Calcif. Tissue Int. 2012; 91: 131-138https://doi.org/10.1007/s00223-012-9618-1
- An accurate estimation of bone density improves the accuracy of subject-specific finite element models.J. Biomech. 2008; 41: 2483-2491https://doi.org/10.1016/j.jbiomech.2008.05.017
- Fiji - an open platform for biological image analysis.Nat. Methods. 2009; 9https://doi.org/10.1038/nmeth.2019.Fiji
- The biomechanics of plate fixation of periprosthetic femoral fractures near the tip of a total hip implant: cables, screws, or both? Proc. Inst.Mech. Eng. Part H J. Eng. Med. 2011; 225: 845-856https://doi.org/10.1177/0954411911413060
- Incidence and predisposing factors of periprosthetic proximal femoral fractures: a literature review.Int. Orthop. 2015; 39: 1673-1682https://doi.org/10.1007/s00264-015-2721-y
- Balance disorder and increased risk of falls in osteoporosis and kyphosis: significance of kyphotic posture and muscle strength.Osteoporos. Int. 2005; 16: 1004-1010https://doi.org/10.1007/s00198-004-1791-2
- Physiologically based boundary conditions in finite element modelling.J. Biomech. 2007; 40: 2318-2323https://doi.org/10.1016/j.jbiomech.2006.10.038
- The material mapping strategy influences the accuracy of CT-based finite element models of bones: an evaluation against experimental measurements.Med. Eng. Phys. 2007; 29: 973-979https://doi.org/10.1016/j.medengphy.2006.10.014
- Elsevicr Science Lrd for IPEMB Stress and strain distribution within the intact femur: compression or bending?.Med. Eng. Hys. 1996; 18: 122-131
- Nonlinear quasi-static finite element simulations predict in vitro strength of human proximal femora assessed in a dynamic sideways fall setup.J. Mech. Behav. Biomed. Mater. 2016; 57: 116-127https://doi.org/10.1016/j.jmbbm.2015.11.026
- Effects of material properties of femoral hip components on bone remodeling.J. Orthop. Res. 1992; 10: 845-853https://doi.org/10.1002/jor.1100100614
- Effects of gait speed of Femoroacetabular joint forces.Appl. Bionics Biomech. 2017; 2017https://doi.org/10.1155/2017/6432969
Published online: October 23, 2022
Accepted: October 14, 2022
Received: July 11, 2022
© 2022 Elsevier Ltd. All rights reserved.