- •Thin, flexible, lightweight distance sensors were placed inside transtibial sockets.
- •Sit, seated shift, stand, standing weight-shift, walk, partial doff, and non-use were detected.
- •Step count data did not well-reflect active use (stands + walks + standing weight-shifts).
- •Participants who had the highest active use temporarily doffed the most often.
Step activity monitors provide insight into the amount of physical activity prosthesis users conduct but not how they use their prosthesis. The purpose of this research was to help fill this void by developing and testing a technology to monitor bodily position and type of activity.
Thin inductive distance sensors were adhered to the insides of sockets of a small group of transtibial prosthesis users, two at proximal locations and two at distal locations. An in-lab structured protocol and a semi-structured out-of-lab protocol were video recorded, and then participants wore the sensing system for up to 7 days. A data processing algorithm was developed to identify sit, seated shift, stand, standing weight-shift, walk, partial doff, and non-use. Sensed distance data from the structured and semi-structured protocols were compared against the video data to characterize accuracy. Bodily positions and activities during take-home testing were tabulated to characterize participants' use of the prosthesis.
Sit and walk detection accuracies were above 95% for all four participants tested. Stand detection accuracy was above 90% for three participants and 62.5% for one participant. The reduced accuracy may have been due to limited stand data from that participant. Step count was not proportional to active use time (sum of stand, walk, and standing weight-shift times).
Step count may provide an incomplete picture of prosthesis use. Larger studies should be pursued to investigate how bodily position and type of activity may facilitate clinical decision-making and improve the lives of people with lower limb amputation.
Abbreviations:ADA (American Disabilities Act), C (Capacitor), ECHO (A data logger used for long-term monitoring), IRB (Institutional Review Board), kΩ (kilohm), L (Inductor), LDC counts (Signal output units for the WAFERs), pf (picofarad), WAFER (A custom flexible printed circuit board antenna used as the sensing element in the distance sensor)
To read this article in full you will need to make a payment
Purchase one-time access:Academic & Personal: 24 hour online accessCorporate R&D Professionals: 24 hour online access
One-time access price info
- For academic or personal research use, select 'Academic and Personal'
- For corporate R&D use, select 'Corporate R&D Professionals'
Subscribe:Subscribe to Clinical Biomechanics
Already a print subscriber? Claim online access
Already an online subscriber? Sign in
Register: Create an account
Institutional Access: Sign in to ScienceDirect
- Prosthetists’ perceptions of information obtained from a lower limb prosthesis monitoring system: a pilot study.J. Prosthet. Orthot. 2019; 31: 112-120https://doi.org/10.1097/JPO.0000000000000203
- Fabricating sockets with distance sensors for prosthesis use and socket fit monitoring.J. Prosthet. Orthot. 2022; (submitted)
- Efficacy of shock-absorbing versus rigid pylons for impact reduction in transtibial amputees based on laboratory, field, and outcome metrics.J. Rehabil. Res. Dev. 2005; 42: 795-808https://doi.org/10.1682/jrrd.2005.02.0034
- Adjustable sockets may improve residual limb fluid volume retention in transtibial prosthesis users.Prosthet. Orthot. Int. 2019; 43: 250-256https://doi.org/10.1177/0309364618820140
- Incorporating a ferrous polymer target into elastomeric liners for socket fit sensing in prosthesis users.Sensors (Basel). 2020; 20: 5620https://doi.org/10.3390/s20195620
- Technology for monitoring everyday prosthesis use: a systematic review.J. Neuroeng. Rehabil. 2020; 17: 93https://doi.org/10.1186/s12984-020-00711-4
- Quantification of prosthetic outcomes: elastomeric gel liner with locking pin suspension versus polyethylene foam liner with neoprene sleeve suspension.J. Rehabil. Res. Dev. 2004; 41: 591-602https://doi.org/10.1682/jrrd.2004.04.0591
- Activity detection using fusion of multi-pressure sensors in insoles.in: 2018 24th International Conference on Pattern Recognition (ICPR), Institute of Electrical and Electronics Engineers, 20–24 Aug 2018. 2018: 3315-3321https://doi.org/10.1109/ICPR.2018.8545834
- Categorisation of activities of daily living of lower limb amputees during short-term use of a portable kinetic recording system: a preliminary study.J. Prosthet. Orthot. 2011; 23: 2-11https://doi.org/10.1097/JPO.0b013e318207914c
- Monitoring prosthesis user activity and doffing using an activity monitor and proximity sensors.J. Prosthet Orthot. 2016; 28: 68-77https://doi.org/10.1097/JPO.0000000000000093
- Physical performance and self-report outcomes associated with use of passive, adaptive, and active prosthetic knees in persons with unilateral, transfemoral amputation: randomized crossover trial.J. Rehabil. Res. Dev. 2015; 52: 677-700https://doi.org/10.1682/JRRD.2014.09.0210
- Considerations for development of sensing and monitoring tools to facilitate treatment and care of persons with lower-limb loss: a review.J. Rehabil. Res. Dev. 2014; 51: 1-14https://doi.org/10.1682/JRRD.2013.01.0024
- Functional assessment and satisfaction of transfemoral amputees with low mobility (FASTK2): a clinical trial of microprocessor-controlled vs. non-microprocessor-controlled knees.Clin. Biomech. 2018; 58 (2018): 116-122https://doi.org/10.1016/j.clinbiomech.2018.07.012
- Vacuum-assisted socket suspension compared with pin suspension for lower extremity amputees: effect on fit, activity, and limb volume.Arch. Phys. Med. Rehabil. 2011; 92: 1570-1575https://doi.org/10.1016/j.apmr.2011.05.019
- Post-doffing residual limb volume change in people with trans-tibial amputation.Prosthetics Orthot. Int. 2012; 36: 443-449https://doi.org/10.1177/0309364612444752
- Duration and frequency of every day activities in total hip patients.J Biomech. 2001; 34: 873-881https://doi.org/10.1016/s0021-9290(01)00035-5
- How do walking, standing, and resting influence trans-tibial amputee residual limb fluid volume?.J. Rehabil. Res. Dev. 2014; 51: 201-212https://doi.org/10.1682/JRRD.2013.04.0085
- Does temporary socket removal affect residual limb fluid volume of trans-tibial amputees?.Prosthetics Orthot. Int. 2016; 40: 320-328https://doi.org/10.1177/0309364614568413
- Daily step count of British military males with bilateral lower limb amputations: a comparison of in-patient rehabilitation with the consecutive leave period between admissions.Prosthetics Orthot. Int. 2019; 43: 188-195https://doi.org/10.1177/0309364618806058
- Instrumented socket inserts for sensing interaction at the limb-socket interface.Med. Eng. Phys. 2018; 51: 111-118https://doi.org/10.1016/j.medengphy.2017.11.006
- Textile-based pressure sensors for monitoring prosthetic-socket interfaces.IEEE Sensors J. 2021; 21: 9413-9422https://doi.org/10.1109/JSEN.2021.3053434
- Thin magnetically permeable targets for inductive sensing: application to limb prosthetics.Sensors. 2019; 19: 4041https://doi.org/10.3390/s19184041
- Automatic control of prosthetic socket size for people with transtibial amputation: implementation and evaluation.IEEE Trans. Biomed. Eng. 2021; 68: 36-46https://doi.org/10.1109/TBME.2020.2992739
- Performance of an auto-adjusting prosthetic socket during walking with intermittent socket release.J. Rehabil. Assist. Technol. 2022; 9: 1-14https://doi.org/10.1177/20556683221093271
- Step activity and 6-minute walk test outcomes when wearing low-activity or high-activity prosthetic feet.Am. J. Phys. Med. Rehabil. 2017; 96: 294-300https://doi.org/10.1097/PHM.0000000000000607
- Effects of activity intensity, time, and intermittent doffing on daily limb fluid volume change in people with transtibial amputation.Prosthetics Orthot. Int. 2019; 43: 28-38https://doi.org/10.1177/0309364618785729
Published online: August 17, 2022
Accepted: August 12, 2022
Received: January 27, 2022
© 2022 Elsevier Ltd. All rights reserved.